This article is a US Government work and is in the public domain in the USA.
Fiber-optic immunosensor for mycotoxins†
Article first published online: 13 DEC 2000
This article is a US government work and is in the public domain in the United States.
Volume 7, Issue 6, pages 371–376, November/December 1999
How to Cite
Maragos, C. M. and Thompson, V. S. (1999), Fiber-optic immunosensor for mycotoxins. Nat. Toxins, 7: 371–376. doi: 10.1002/1522-7189(199911/12)7:6<371::AID-NT86>3.0.CO;2-8
- Issue published online: 13 DEC 2000
- Article first published online: 13 DEC 2000
- Manuscript Accepted: 20 JUL 2000
- Manuscript Received: 3 NOV 1999
- fiber optic immunosensor;
Evanescent wave-based fiber-optic immunosensors were studied for the detection of fumonisins and aflatoxins in maize. Two formats, competitive and non-competitive, were used. A competitive format was used to measure fumonisin B1 (FB1) in both spiked and naturally contaminated maize samples. Fumonisin monoclonal antibodies were covalently coupled to an optical fiber and the competition between FB1 and FB1 labeled with fluorescein (FB1-FITC) for the limited number of binding sites on the fiber was assessed. The signal generated in the assay was inversely proportional to the FB1 concentration. For samples, the concentration causing an inhibition of binding by 50 % (IC50) was dependent upon the clean-up procedure used. Simple dilution of methanolic maize extracts yielded an assay with an IC50 equivalent to 25 µg FB1 g−1 maize with a limit of detection of 3.2 µg g−1 maize. Affinity column clean-up yielded an assay with an IC50 equivalent to 5 µg FB1 g−1 maize (limit of detection 0.4 µg FB1 g−1). An HPLC method and the immunosensor method agreed well for naturally contaminated maize samples except when large amounts of other fumonisins that cross-react with the immunosensor were present. The second sensor format, for the mycotoxin aflatoxin B1 (AFB1), was a non-competitive assay using the native fluorescence of this mycotoxin. Because the fluorescence of AFB1 itself was detected, the response of the sensor was directly proportional to the toxin concentration. The sensor, while capable of detecting as little as 2 ng ml−1 of AFB1 in solution was technically not an immunosensor, since the attachment of aflatoxin specific antibodies was not required. Sensors of the formats described have the potential to rapidly screen individual maize samples but require coupling with a clean-up technique to be truly effective. Published in 1999 by John Wiley & Sons, Ltd.