Single-laboratory validation of a method to quantify bound 2-chloropropane-1,3-diol and 3-chloropropane-1,2-diol in foodstuffs using acid catalysed transesterification, HFBI derivatisation and GC/MS detection
Version of Record online: 14 FEB 2011
Copyright © 2011 WILEY-VCH Verlag GmbH & Co. KGaA, Weinheim
European Journal of Lipid Science and Technology
Special Issue: Fatty acid esters of chloropropanols and glycidol
Volume 113, Issue 3, pages 345–355, March 2011
How to Cite
Hamlet, C. G. and Asuncion, L. (2011), Single-laboratory validation of a method to quantify bound 2-chloropropane-1,3-diol and 3-chloropropane-1,2-diol in foodstuffs using acid catalysed transesterification, HFBI derivatisation and GC/MS detection. Eur. J. Lipid Sci. Technol., 113: 345–355. doi: 10.1002/ejlt.201000492
- Issue online: 10 MAR 2011
- Version of Record online: 14 FEB 2011
- Accepted manuscript online: 19 JAN 2011 05:10AM EST
- Manuscript Accepted: 23 DEC 2010
- Manuscript Revised: 9 DEC 2010
- Manuscript Received: 7 OCT 2010
A protocol for the measurement of bound 2-chloropropane-1,3-diol (2-CPD) and 3-chloropropane-1,2-diol 3-CPD) in foodstuffs using acid catalysed transesterification and GC/MS/MS has undergone single-laboratory validation. Validation of the method included an assessment of recovery (bias) of 3-CPD from reference 3-CPD mono- and diesters using a range of fortified and unfortified cereal, oil and potato products and participation in a EC proficiency test on 3-CPD-esters. Method repeatability was assessed by measuring bound 2- and 3-CPD present in a processed cereal product. The results indicated that the method was fit for purpose for the measurement of bound 2- and 3-CPD over concentration ranges of 40–10 000 µg/kg for oils, and 4–4000 µg/kg for foodstuffs depending on the fat content. The selectivity/robustness of the method was demonstrated by fortifying samples with ‘free’ 3-CPD, chloride ions, glycerol and glycidyl butyrate: the procedure was found to be selective for ‘bound’ 3-CPD and no adventitious formation of 2- or 3-CPD occurred during analysis.
Practical applications: This method provides a single-laboratory validation of a method to measure amounts of bound 2- and 3-CPD in a wide range food. The method provides the information necessary to assist researchers engaged in the selection and validation of analytical methods to measure these contaminants in foods.