Continuous measurement of nitrate concentration in a highly event-responsive agricultural catchment in south-west of France: is the gain of information useful?
Version of Record online: 11 MAY 2012
Copyright © 2012 John Wiley & Sons, Ltd.
Volume 27, Issue 12, pages 1751–1763, 15 June 2013
How to Cite
Ferrant, S., Laplanche, C., Durbe, G., Probst, A., Dugast, P., Durand, P., Sanchez-Perez, J. M. and Probst, J. L. (2013), Continuous measurement of nitrate concentration in a highly event-responsive agricultural catchment in south-west of France: is the gain of information useful?. Hydrol. Process., 27: 1751–1763. doi: 10.1002/hyp.9324
- Issue online: 4 JUN 2013
- Version of Record online: 11 MAY 2012
- Accepted manuscript online: 4 APR 2012 11:27AM EST
- Manuscript Accepted: 6 MAR 2012
- Manuscript Received: 1 AUG 2010
- agricultural nitrate;
- storm chasing protocol;
- nitrate flushing
A nitrate sensor has been set up to measure every 10 min the nitrate signal in a stream draining a small agricultural catchment dominated by fertilized crops during a 2-year study period (2006–2008) in the south-west of France. An in situ sampling protocol using automatic sampler to monitor flood events have been used to assume a point-to-point calibration of the sensor values. The nitrate concentration exhibits nonsystematic concentration and dilution effects during flood events. We demonstrate that the calibrated nitrate sensor signal gathered from the outlet is considered to be a continuous signal using the Nyquist–Shannon sampling theorem. The objectives of this study are to quantify the errors generated by a typical infrequent sampling protocol and to design appropriate sampling strategy according to the sampling objectives. Nitrate concentration signal and flow data are numerically sampled to simulate common sampling frequencies. The total fluxes calculated from the simulated samples are compared with the reference value computed on the continuous signal. Uncertainties are increasing as sampling intervals increase; the method that is not using continuous discharge to compute nitrate fluxes bring larger uncertainty. The dispersion and bias computed for each sampling interval are used to evaluate the uncertainty during each hydrological period. High underestimation is made during flood periods when high-concentration period is overlooked. On the contrary, high sampling frequencies (from 3 h to 1 day) lead to a systematic overestimation (bias around 3%): highest concentrations are overweighted by the interpolation of the concentration in such case. The in situ sampling protocol generates less than 1% of load estimation error and sample highest concentration peaks. We consider useful such newly emerging field technologies to assess short-term variations of water quality parameters, to minimize the number of samples to be analysed and to assess the quality state of the stream at any time. Copyright © 2012 John Wiley & Sons, Ltd.