The analysis of 15N in aqueous samples requires the concentration of dissolved nitrogen (N) into a small volume that can be analysed by mass spectrometry. This is conveniently achieved by the NH3 diffusion technique, where NH is captured on small acidified filters enclosed in PTFE. NO can be analysed the same way by reducing it to NH with Devarda's alloy. H2SO4 is commonly used for the acidification of the filters. During combustion, however, this acid leads to the production of SO2 and elemental sulphur, which both have detrimental effects on the mass spectrometer. We propose here to replace H2SO4 with citric acid because it is combusted completely to CO2 and H2O in the elemental analyser before entering the mass spectrometer. Citric acid was found to give comparable results in terms of N recovery and 15N values, both for NH and for NO samples. Blank samples revealed that N contamination was slightly lower using citric instead of sulphuric acid as acidifier of the glass filters. NH samples first concentrated over cation-exchange columns were strongly acidic and several methods were tested to raise the pH for the subsequent diffusion. These samples gave incomplete N recoveries, but this problem was independent of the acid used on the filters and of the final pH of the sample. Complete recovery was achieved only by increasing the volume of the eluate from the columns. Citric acid can thus generally be recommended instead of H2SO4 for ammonia diffusion. Copyright © 2006 John Wiley & Sons, Ltd.
If you can't find a tool you're looking for, please click the link at the top of the page to "Go to old article view". Alternatively, view our Knowledge Base articles for additional help. Your feedback is important to us, so please let us know if you have comments or ideas for improvement.