A response to a previously published article or letter can be submitted to the Online Correspondence section at http://www.anaesthesiacorrespondence.com. Please note that a selection of this correspondence will be reproduced (possibly in modified form) in the journal. All correspondence intended for publication in Anaesthesia should be addressed to Dr Steve Yentis, Editor-in-Chief, and submitted as an email attachment to email@example.com. Copy should be prepared in the usual style of the Correspondence section. Authors must follow the advice about references and other matters contained in the Guidance for Authors at http://wileyonlinelibrary.com/journal/anae. Correspondence presented in any other style or format will be returned to the author for revision. All correspondence submissions should be accompanied by a completed Author Declaration Form which can be accessed via a link under ‘Covering letter’ in the Guidance for Authors (as above). The completed Author Declaration Form should be sent either by e-mail with the submission or by fax to (0)207 681 1008.
We routinely perform bronchoscopy following percutaneous tracheostomy to check if the carina is visible and to aspirate blood and secretions. During a recent case, we found a section of plastic tubing lodged in a patient’s right main bronchus following tracheostomy. We initially suspected that it had originated from the tracheal dilator, as we were using new equipment with which we had less experience. Upon retrieval, we discovered that it was the end of a suction catheter from a closed suction device (Kimberly-Clark Ltd, West Malling, Kent, UK, Fig. 3). Close inspection revealed that it had been cut cleanly, as if with a pair of scissors.
Further enquiry revealed that the previous tracheal tube had been cut in situ because the suction catheter would not pass easily along that. We concluded that the suction catheter had not been pulled back fully and that the end had been accidentally severed as it still lay inside the tracheal tube when the latter was cut. Retrospective inspection of chest radiographs showed no signs of a foreign body.
Cutting tracheal tubes in situ is hazardous, as it may prove impossible to reinsert the tracheal tube connector into the new cut end of the tube. Our case provides a further reason to apply caution when performing this manoeuvre.
No external funding and no competing interests declared.