A response to a previously published article or letter can be submitted to the Online Correspondence section at http://www.anaesthesiacorrespondence.com. Please note that a selection of this correspondence will be reproduced (possibly in modified form) in the journal. All correspondence intended for publication in Anaesthesia should be addressed to Dr Steve Yentis, Editor-in-Chief, and submitted as an email attachment to firstname.lastname@example.org. Copy should be prepared in the usual style of the Correspondence section. Authors must follow the advice about references and other matters contained in the Guidance for Authors at http://wileyonlinelibrary.com/journal/anae. Correspondence presented in any other style or format will be returned to the author for revision. All correspondence submissions should be accompanied by a completed Author Declaration Form which can be accessed via a link under ‘Covering letter’ in the Guidance for Authors (as above). The completed Author Declaration Form should be sent either by e-mail with the submission or by fax to (0)207 681 1008.
Plastic wrappers: not just an environmental hazard
Version of Record online: 14 MAR 2011
Anaesthesia © 2011 The Association of Anaesthetists of Great Britain and Ireland
Volume 66, Issue 4, page 321, April 2011
How to Cite
Smith, T., Clements, L. and Plaat, F. (2011), Plastic wrappers: not just an environmental hazard. Anaesthesia, 66: 321. doi: 10.1111/j.1365-2044.2011.06673.x
Visit the Anaesthesia Correspondence website at http://www.anaesthesiacorrespondence.com and comment on any article or letter in this issue of the Journal.
- Issue online: 14 MAR 2011
- Version of Record online: 14 MAR 2011
We would like to report a near miss that we believe is likely to recur. During an elective operating list, as we inserted the laryngoscope blade (Truphatek International Ltd, Netanya, Israel) into the mouth of an anaesthetised patient, our vigilant anaesthetic nurse, who was positioned close to the head of the patient, suddenly lunged dramatically for the scope. She had noticed a nearly invisible scrap of clear plastic on the tip of the laryngoscope blade (Fig. 6). Although we routinely check the laryngoscope blade to ensure that the light is working, we had not seen the piece of plastic as it was attached to the inner aspect of the blade. Our experience is not unique. A search of the published literature has revealed a number of similar incidents [1, 2] and there has been a recent harrowing report of a clinical near miss involving a small disc of plastic wrapper causing complete obstruction of the anaesthetic circuit .
Given that every component of the anaesthetic breathing system is wrapped in clear plastic packaging (Fig. 7), it is not surprising that similar incidents continue to occur. The risk is reduced if the package is opened along the perforations, but frequently these are poorly designed and it is easier to tear the equipment out of the wrapper. We have reported the incident to the manufacturer, the regulatory agent for the device and the National Patient Safety Agency. One way of reducing the risk would be through the use of coloured, semi-translucent wrapping material and the manufacturer has agreed to investigate this.
No external funding and no competing interests declared.
- 3Anon. Breathing system obstruction: a cautionary tale. Anaesthesia News 2010; 278: 8.