Fifty-seventh annual meeting of the American association of physicists in medicine
SU-E-P-46: Clinical Acceptance Testing and Implementation of a Portable CT Unit
Planning for the first installation in New England of a new portable CT unit to be used in the Operating Room required the integration of many departments including Surgery, Neurosurgery, Information Services, Clinical Engineering, Radiology and Medical Physics/Radiation Safety. Acceptance testing and the quality assurance procedures were designed to optimize image quality and patient and personnel radiation exposure.
The vendor's protocols were tested using the CT Dosimetry phantoms. The system displayed the CTDIw instead of the CTDIvol while testing the unit. Radiation exposure was compared to existing CT scanners from installed CT units throughout the facility. Brainlab measures all 4 periphery slots on the CT Dosimetry phantom. The ACR measures only the superior slot for the periphery measurement. A comprehensive radiation survey was also performed for several locations.
The CTDIvol measurements were comparable for the following studies: brain, C-Spine, and sinuses. However, the mobile CT measurements were slightly higher than other CT units but within acceptable tolerance if measured using the ACR method.Based on scatter measurements, it was determined if any personnel were to stay in the OR Suite during image acquisition that the appropriate lead apron and thyroid shields had to be worn.In addition, to reduce unnecessary scatter, there were two mobile 6 foot wide shields (1/16″ lead equivalent) available to protect personnel in the room and adjacent areas.
Intraoperative CT provides the physician new opportunities for evaluation of the progression of surgical resections and device placement at the cost of increasing the amount of trained personnel required to perform this procedure. It also brings with it challenges to keep the radiation exposure to the patients and staff within reasonable limits.