Fifty-eighth annual meeting of the american association of physicists in medicine
SU-F-P-32: A Phantom Study of Accuracy of Four-Dimensional Cone-Beam CT (4D-CBCT) Vs. Three-Dimensional Cone Beam CT (3D-CBCT) in Image Guided Radiotherapy
SymmetryTM 4D IGRT system of Elekta has been installed at our institution, which offers the 4D CBCT registration option. This study is to evaluate the accuracy of 4D CBCT system by using the CIRS 4D motion phantom and to perform a feasibility study on the implementation of 4D-CBCT as image guidance for SBRT treatment.
The 3D and 4D CT image data sets are acquired using the CIRS motion phantom on a Philips large bore CT simulator. The motion was set as 0.5 cm superior and inferior directions with 6 seconds recycle time. The 4D CT data were sorted as 10 phases. One identifiable part of the 4D CT QA insert from CIRS phantom was used as the target. The ITV MIP was drawn based on maximum intensity projection (MIP) and transferred as a planning structure into 4D CBCT system. Then the 3D CBCT and 4D CBCT images were taken and registered with the free breath (3D), MIP (4D) and average intensity projection (AIP)(4D) reference data sets. The couch shifts (X, Y, Z) are recorded and compared.
Table 1 listed the twelve couch shifts based on the registration of MIP, AIP and free breath CT data sets with 3D CBCT and 4D CBCT for both whole body and local registration. X, Y and Z represent couch shifts in the direction of the right-left, superior-inferior and anterior-posterior. The biggest differences of 0.73 cm and 0.57 cm are noted in the free breath CT data with 4D CBCT and 3D CBCT data registration. Fig. 1 and Fig. 2 are the shift analysis in diagram. Fig. 3 shows the registration.
Significant differences exist in the shifts corresponding with the direction of target motion. Further investigations are ongoing.