Fifty-eighth annual meeting of the american association of physicists in medicine
SU-F-P-40: Analysis of Pelvic Lymph Node Margin Using Prostate Fiducial Markers, for SBRT Treatments
The use of fiducials markers in prostate treatment allows a precise localization of this volume. Typical prostate SBRT margins with fiducials markers are 5mm in all directions, except toward the rectum, where 3mm is used. For some patients nearby pelvic lymph nodes with 5mm margin need to be irradiate assuming that its localization is linked to the prostate fiducial markers instead of bony anatomy. The purpose of this work was to analyze the geometric impact of locate the lymph node regions through the patient positioning by prostate fiducial markers.
10 patients with prostate SBRT with lymph nodes irradiation were selected. Each patient had 5 implanted titanium fiducial markers. A Novalis TX (BrainLAB-Varian) with ExacTrac and aSi1000 portal image was used. Treatment plan uses 11 beams with a dose prescription (D95%) of 40Gy to the prostate and 25Gy to the lymph node in 5 fractions. Daily positioning was carried out by ExacTrac system based on the implanted fiducials as the reference treatment position; further position verification was performed using the ExacTrac and two portal images (gantry angle 0 and 90) based on bony structures. Comparison between reference position with bony based ExacTrac and portal image localization, was done for each treatment fraction
A total of 50 positioning analysis were done. The average discrepancy between reference treatment position and ExacTrac based on bony anatomy (pubic area) was 4.2mm [0.3; 11.2]. The discrepancy was <5mm in 61% of the cases and <9mm in 92%. Using portal images the average discrepancy was 3.7mm [0.0; 11.1]. The discrepancy was <5mm in 69% of the cases and <9mm in 96%.
Localizing lymph node by prostate fiducial markers may produce large discrepancy as large as 11mm compared to bony based localization. Dosimetric impact of this discrepancy should be studied.