Accelerated and quantitative three-dimensional molecular MRI using a generative adversarial network

Purpose: To substantially shorten the acquisition time required for quantitative three-dimensional (3D) chemical exchange saturation transfer (CEST) and semisolid magnetization transfer (MT) imaging and allow for rapid chemical exchange parameter map reconstruction. Methods: Three-dimensional CEST and MT magnetic resonance fingerprinting (MRF) datasets of L-arginine phantoms, whole-brains, and calf muscles from healthy volunteers, cancer patients, and cardiac patients were acquired using 3T clinical scanners at three different sites, using three different scanner models and coils. A saturation transfer-oriented generative adversarial network (GAN-ST) supervised framework was then designed and trained to learn the mapping from a reduced input data space to the quantitative exchange parameter space, while preserving perceptual and quantitative content. Results: The GAN-ST 3D acquisition time was 42–52 s, 70% shorter than CEST-MRF. The quantitative reconstruction of the entire brain took 0.8 s. An excellent agreement was observed between the ground truth and GAN-based L-arginine concentration and pH values (Pearson’s r > 0.95, ICC > 0.88, NRMSE < 3%). GAN-ST images from a brain-tumor subject yielded a semi-solid volume fraction and exchange rate NRMSE of 3.8 ± 1.3% and 4.6 ± 1.3%, respectively, and SSIM of 96.3 ± 1.6% and 95.0 ± 2.4%, respectively. The mapping of the calf-muscle exchange parameters in a cardiac patient, yielded NRMSE < 7% and SSIM > 94% for the semi-solid exchange parameters. In regions with large susceptibility artifacts, GAN-ST has demonstrated improved performance and reduced noise compared to MRF. Conclusion: GAN-ST can substantially reduce the acquisition time for quantitative semi-solid MT/CEST mapping, while retaining performance even when facing pathologies and scanner models that were not available during training.


I N T RO DUCTI ON
Semi-solid MT and chemical exchange saturation transfer (CEST) are MRI techniques that provide unique contrast, based on saturation transfer (ST).While semi-solid MT provides a means for studying macromolecules, lipids, and myelin, 1 CEST enables the detection of millimolar concentrations of mobile proteins, peptides, and metabolites. 2However, most CEST imaging protocols provide a semi-quantitative contrast by way of the magnetization transfer ratio asymmetry (MTR asym ) analysis.While this metric has shown clear value in a variety of clinical studies, [3][4][5] it is dependent on numerous factors such as chemical exchange rate, volume fraction of the exchangeable solute protons, water longitudinal relaxation rate, radiofrequency (RF) saturation time, radiofrequency saturation power, and water transverse relaxation rate, all of which must be modeled across a variety of exchangeable proton pools. 4,6Additionally, while clinical amide proton transfer weighted CEST protocols can now be acquired in just a few minutes, 7 the full Z-spectrum acquisition required for the analysis of multiple exchangeable proton pools (e.g., via Lorentzian model fitting) requires significantly longer acquisition times. 8In addition, for quantification of the exchange parameters, the acquisition of multiple Z-spectra with different saturation powers is typically needed, 9 resulting in even longer acquisition times.
A variety of methods were previously developed for accelerating ST-weighted MRI. 10 Prominent examples include parallel imaging and transmission, 11 compressed sensing 12 ultrafast Z-spectroscopy, 13 and pulsed steady-state CEST sequences. 14Recently, rapid developments in deep learning have been harnessed for semisolid MT/CEST imaging.Fully connected neural networks have been used to rapidly extract Lorentzian fitted parameters 15 and apparent exchange dependent-relaxation parameters from Z-spectra 16 and to predict 9.4T CEST contrast from 3T data. 17Convolutional neural networks have been used for B 0 correction, 18 SNR enhancement, 19 tumor classification, 20 assessment of tumor progression, 21 and eightfold acceleration of CEST-weighted image acquisition. 22agnetic resonance fingerprinting (MRF) 23 is a rapid and quantitative imaging paradigm that uses pseudo-random acquisition schedules to acquire unique signal trajectories, which are then matched to an existing database or "dictionary" of Bloch-equation-based simulated signals, providing a de facto pixelwise estimation of the underlying magnetic properties.Although initially developed for quantification of water T 1 and T 2 relaxation times, MRF has recently been modified and expanded for the quantification of semi-solid MT and CEST exchange parameters, 6,24,25 in an attempt to ameliorate the above-mentioned challenges of conventional ST-weighted imaging methods.
In many pathologies, the chemical exchange parameters of multiple proton pools (e.g.MT and amide) vary simultaneously and must be included in any MRF simulated dictionary.This leads to an exponential growth in dictionary size and hence very long dictionary generation and parameter matching times.7][28][29][30][31] However, due to the larger number of tissue parameters that must be matched for CEST/MT fingerprinting, the implementation of neural networks for exchange parameter map reconstruction is more complicated than that employed for "conventional" T 1 /T 2 MRF, and require a separate optimization and research endeavor. 32,33ccordingly, several deep-learning-based approaches have been developed specifically for shortening the reconstruction part of the semisolid MT/CEST MRF imaging pipeline. 34,35While these strategies have demonstrated promising results, the acquisition time is still long and requires the acquisition of T 1 , T 2 , and B 0 maps, constituting an obstacle for routine clinical adoption. 32Moreover, applying CEST-MRF for multislice imaging would further increase the scan time.
Recently, a unique machine learning approach has demonstrated the ability to learn the hidden and complicated relations (manifold) between two paired image categories 36 and generate an approximation of the appropriate image-pair for a given input.This approach, termed conditional generative adversarial network (GAN), 37 is built on two competing neural networks, a generator and a discriminator, which are trained simultaneously.The generator aims to synthesize convincingly realistic samples while the discriminator estimates the probability that a sample comes from the training data category. 38During training, the generator gradually learns to create more convincing models based on the discriminator's feedback.The GAN framework is highly modular, and adversarial models have been shown to be effective for a variety of applications.In particular, the conditional GAN "pix2pix" architecture was developed for image-to-image translation problems. 37Conditional GANs have shown great performance in translational tasks involving natural images, such as the synthesis of night views from pictures taken at daytime, generation of full object photos from edges, and maps from aerial photographs.The promise of conditional GANs has recently been translated and expanded into medical imaging, where this strategy was employed for cross modality synthesis (e.g., MRI to CT) and transformation between T 1 and T 2 -weighted MRI maps.Here, we hypothesized, that a modified conditional GAN framework could be designed and trained to learn the manifold that links between raw semisolid MT/CEST-MRF encoded images and their quantitative exchange parameter image counterparts.Moreover, we assumed that an efficient quantification could still be obtained as the number of raw MRF encoded images is reduced, thereby allowing a substantial shortening of the acquisition time.Finally, to transform the developed approach into a clinically attractive and practical protocol, we have combined the MRF acquisition block with a three-dimensional (3D) Snapshot CEST readout module, 40 allowing rapid whole-brain (or any other organ) multi-slice imaging.

ST-oriented GAN (GAN-ST) architecture
A supervised learning framework (Figure 1) was designed based on the conditional GAN architecture. 37The generator was a U-Net convolutional network aiming to synthesize two proton exchange parameter maps (volume fraction and exchange rate), for either the semisolid MT or the CEST compound exchangeable protons.The discriminator aimed to predict whether the images are the "real' corresponding quantitative images, or a "fake" (generator synthesized maps).The ground truth was obtained by a dictionary-trained fully connected semisolid MT/CEST-MRF neural-network (Figure 1B) that received the full MRF acquisition schedule as input (M = 30 raw images).GAN-ST was trained to yield the same quantitative maps by receiving only a partial subset of N = 9 raw MRF encoding images as input.For the human brain imaging scenario, the water T 1 , T 2 , and B 0 may vary significantly in the WM/GM/tumor tissues.Thus, to improve the reliability and accuracy of the ground-truth for this case, the T 1 , T 2 , and B 0 maps were acquired separately, quantified, and given as an additional direct input to the reference ground truth MT/CEST-MRF network, as performed and described by Perlman et al. 34 Notably, these three maps were not given to GAN-ST.To preserve the perceptual and quantitative content of the original quantitative images while retaining spatial continuity and Generative adversarial network (GAN)-saturation transfer (ST) architecture.(A) A conditional GAN framework, receives N raw, molecular information encoding semi-solid magnetization transfer/chemical exchange saturation transfer images, and is trained to simultaneously output the quantitative proton volume fraction and the exchange rate maps.(B) A fully connected neural network, receiving the full-length raw magnetic resonance fingerprinting image series (M > N) pixelwise, as well as T 1 , T 2 , and B 0 maps, and yielding the reference proton volume fraction and exchange rate maps. 34The output of this network was used for training GAN-ST.

T A B L E 1
Human imaging data cohort smoothness in the GAN-ST output, the following total loss (L total ) function was used: Where L 1 is the pixelwise l 1 content loss, L adv is the adversarial loss determined by the discriminator, L tv is the total variation loss, and L p is the perceptual loss. 41The latter was defined as the l 2 loss between the feature maps obtained from activating a pretrained deep convolutional network architecture (VGG19 42 ) on the ground truth and on the GAN-ST approximated quantitative maps. 1,2,3,4 are the loss weights, determined using a separate validation image set (Table 1).The training was performed for 400 epochs with a batch size of 4. The learning rates of the generator and discriminator were 0.0001 and 0.0005, respectively.The method was realized in Keras 43 and implemented on a desktop computer equipped with a single Nvidia GeForce RTX 3080 GPU.

CEST phantoms
A set of L-arginine (L-arg, chemical shift = 3 ppm) phantoms was prepared by dissolving L-arg (Sigma-Aldrich) in a pH 4 Buffer, at a concentration of 25, 50, or 100 mM.The phantoms were titrated to different pH levels between 4.0-6.0,and placed in a dedicated holder, inside a container filled with saline/PBS.The phantoms were 3D scanned twenty times at two different imaging sites (Tubingen and Boston) using two scanner models (Prisma and Skyra, Siemens Healthineers, respectively).At each scan, a different subset of 6-7 vials was used from the general range of 4-6 pH with an L-arg concentration of 25, 50, or 100 mM.The imaging was repeated after physically and randomly rotating/moving the phantom.The phantoms were independently prepared at each site.

Human imaging subjects
All in vivo measurements were performed under approval by the local ethics/IRB committee.Each subject gave written, informed consent before the study.A total of 17 subjects were imaged and allocated into the training, validation, or test set, as described in Table 1.The subjects were scanned at three imaging sites (Tubingen, Boston, and Erlangen) and were either healthy volunteers, glioblastoma (GBM) patients, or cardiac patients.The separate validation set was used for setting the hyperparameters (e.g., the number of epochs, as determined by early stopping), and the same determined training parameters were used for all imaging scenarios (phantom, brain, and leg).
The test set was designed to impose a challenging evaluation environment, aiming to explore the GAN-ST robustness and ability to extrapolate beyond the environment of the training samples.In particular, while all training subjects were healthy volunteers, the test set included a GBM patient, a cardiac patient, and a healthy volunteer imaged at a different site and scanner model, which were not used in the training set.

MRI acquisition
The MRI experiments were conducted at three imaging sites using four 3T clinical scanners consisting of three different models (2×Prisma, Trio, and Skyra scanners, Siemens Healthineers) and three coil types (64-channel head coil, 32 channel head coil, and a single-element leg coil).All acquisition schedules were implemented using the Pulseq prototyping framework 44 and the open-source Pulseq-CEST sequence standard. 45The MRF protocol generated M = 30 raw, molecular information encoding images, using a spin lock saturation train (13 × 100 ms, 50% duty-cycle), which varied the saturation pulse power between 0 and 4 T (average pulse amplitude, the complete and exact saturation pulse train parameters are provided in the data availability statement). 34The saturation pulse frequency offset was fixed at 3 ppm for L-arginine phantom imaging 6,33 or varied between 6 and 14 ppm for semisolid MT brain/leg imaging. 34The saturation block was fused with the 3D centric reordered EPI readout module described by Mueler et al. 40 and Akbey et al. 46

In vivo preprocessing
All images were motion-corrected and registered using elastix. 48Gray-matter and white-matter segmentation was performed using statistical parameter mapping (SPM) 49 from a T 1 map.Quantitative reference CEST-MRF maps were obtained using a fully connected neural network trained on simulated dictionaries, where all M = 30 raw input measurements were taken as input.For brain imaging, pixelwise T 1 , T 2 , and B 0 values were also incorporated as direct inputs to the NN.For a detailed description of the CEST-MRF reconstruction and quantification procedure see the recent publication by Perlman et al. 34

Statistical analysis
Pearson's correlation coefficients were calculated using the open-source SciPy scientific computing library for Python. 50Intraclass Correlation Coefficients (ICC) were calculated using the open-source Pingouin statistical package for Python. 51The structural similarity index (SSIM) 52 was computed using the SSIM-python imaging library (PIL).In all box plots, the central horizontal lines represent median values, box limits represent upper (third) and lower (first) quartiles, whiskers represent 1.5 × the interquartile range above and below the upper and lower quartiles, and circles represent outliers.Statistics in the text are presented as mean ± SD.Differences were considered significant at p < 0.05.

Phantom study: direct estimation of concentration and pH
To explore the GAN-ST ability for direct estimation of the pH and compound concentration, we created another set of ground-truth reference images, where all pixels in each segmented vial were replaced by the pH-meter measured pH, and the analytic-scale determined L-arg concentration.GAN-ST was retrained while employing these images as the target, followed by an estimation of the pH and L-arg concentration in a different phantom test set (Figure 2E-H).GAN-ST generated concentration maps were in good agreement with measured values, yielding an NRMSE of 2.7 ± 0.4% and SSIM of 0.889±0.045(Figure 3G,H) and a significant correlation across all slices (Pearson's r = 0.950, ICC = 0.883, p < 0.0001, Figure 3E).Similarly, GAN-ST generated pH maps were in good agreement with measured values, yielding an NRMSE of 0.6 ± 0.3% and SSIM of 0.993 ± 0.006 (Figure 3G,H  significant correlation across all slices (Pearson's r = 0.998, ICC = 0.860, p < 0.0001) (Figure 3F).

In vivo study-brain parameter quantification
A comparison between the GAN-ST results with the number of raw molecular encoding images set to N = 9 and the full-length MRF-based reference (M = 30) for four representative slices is shown in Figure 4 (healthy volunteer scanned at a site and scanner model that were not available in the training cohort) and Figure 5 (tumor patient).The accelerated GAN-ST output was visually very similar to the CEST-MRF reference (average SSIM > 0.925, average NRMSE < 5.2%, Figure 6C,D,G,H).Moreover, in regions with large susceptibility artifacts, GAN-ST demonstrated improved performance and reduced noise compared to MRF (red arrows in Figures 4 and 5).Although the training cohort included only healthy volunteers, GAN-ST was able to output CEST-MRF comparable parameter maps, even in complex tumor and edema containing image slices (Figure 5).The resulting WM/GM f ss estimated by the GAN-ST approach for all nontumor-containing slices was 18.7 ± 2.1 / 13.2 ± 2.5%, compared to 18.7 ± 2.0 / 12.4 ± 2.7% using the full-length MRF.The WM/GM k ssw estimated by the GAN-ST approach was 36.2 ± 6.1 / 51.6 ± 8.5 Hz, compared to 33.9 ± 5.2 / 49.1 ± 8.5 Hz, by the full-length MRF reference, with significant correlation between individual pixel values obtained by both methods (Pearson's r = 0.90 and 0.75, ICC = 0.88 and 0.72, for f ss and k ssw , respectively, p < 0.001, Figure 6A,B).

In vivo study-calf-muscle parameter quantification
The GAN-ST-based in vivo calf exchange parameter quantification (Figure 7) was characterized by an SSIM > 0.94 and an average NRMSE < 7% (Figure 8).A significant correlation was observed between the GAN-predicted and reference CEST-MRF-based parameters, although the semi-solid volume fraction proton quantification was in better agreement with CEST-MRF than the exchange rate (r = 0.73, ICC = 0.71, p < 0.001, and r = 0.51, ICC = 0.45, p < 0.001, respectively).

D I S C U S SION
In recent years, the CEST contrast mechanism has been increasingly studied for a variety of medical applications. 5he molecular information provided by CEST, most commonly by the amide proton transfer (APT) effect, has provided added clinical value compared to traditional T 1 /T 2 -weighted imaging.For example, endogenous CEST signals have been correlated with tumor lesion enhancement following Gadolinium injection. 4,53Furthermore, CEST-weighted images were able to better discriminate treatment related changes from tumor progression. 54MT signals stemming from semi-solid macromolecules have also been shown to be beneficial for cancer characterization and monitoring, either as a standalone method 55 or in combination with CEST. 56In addition, MT has demonstrated potential as a biomarker for pathological skeletal muscle 57 and has long been known for its importance for multiple sclerosis imaging. 58However, both CEST and semi-solid MT imaging are highly sensitive to the acquisition parameters used, as well as to changes in water pool relaxation.Moreover, these methods are prone to bias stemming from the analysis metric used and are subjected to contaminations from the signals originating from other tissue metabolites and compounds. 59All the aforementioned challenges have motivated the development of quantitative approaches for semi-solid MT/CEST imaging. 60,61In the context of clinical imaging, it is clearly important to also strive for short acquisition times.Semisolid MT/CEST MRF has recently been suggested as a rapid and quantitative molecular imaging pipeline. 32However, as the multiprotocol conventional clinical routine is already lengthy (e.g. are also acquired based on the same 3D readout).(2) It employed GANs to further accelerate the 3D molecular scan time by reducing the required number of signal trajectory acquisitions by 70%, requiring acquisition times of less than 1 min.(3) The GAN-ST images have demonstrated the ability to mitigate the noise arising from field inhomogeneity and susceptibility artifacts (e.g., near the sinuses and the eyes, Figures 4 and 5).(4) The GAN-ST was able to extrapolate beyond the training data properties, as demonstrated using a subject scanned at a site and scanner model that were not used for training (Figures 4 and 6E-H).
In terms of the reconstruction (or parameter quantification from raw MRF images), one of the main limitations of "classical fingerprinting," is the lengthy dictionary matching times, where millions of simulated signals need to be compared to the experimental images (e.g., by means of a pixelwise dot-product matching).The framework presented here allows the reconstruction of quantitative proton volume fraction and exchange rate maps of the entire 3D volume acquired, in less than a second.Pixelwise matching using fully connected neural networks has previously been used for rapid semisolid MT/CEST MRF reconstruction. 34However, the use of the GAN architecture, with its inherent spatial dependencies and U-Net structure, has also provided the ability to mitigate susceptibility artifacts (Figures 4 and 5).
The semi-solid proton exchange rate was generally slower in the GAN-ST-based images, compared to CEST-MRF, in the tumor regions (Figure 5I-P), as well as in some anatomical parts of the calf (Figure 7I-K).This is in line with previous quantification attempts that faced similar challenges and reduced accuracy with the semisolid MT exchange rate. 35,62These reports have attributed the poor discrimination ability for this property to the relatively small signal fluctuations that were observed as the exchange rate was varied.In this work, the reduced accuracy can also be explained by the intentionally challenging datasets design, where tumor data were not provided during the training.This assumption is supported by the increased correlation observed between the CEST-MRF and GAN-ST exchange rates for a healthy volunteer that was evaluated (Figure 4, and Figure 6F compared to Figure 6B).The phantom data quantitative exchange parameter maps obtained using GAN-ST with 70% acceleration were in excellent agreement with the CEST-MRF-based reference (Figures 2 and 3).Moreover, the same GAN architecture has allowed the direct estimation of the pH and compound concentration, measured r = 0.77, ICC = 0.74 (p < 0.001) using 20 input images.While the quantitative maps obtained with a drastic acceleration of 70% are deemed both visually and quantitatively satisfactory, future expanded clinical evaluations could determine more accurately the degree of acceleration level sufficient for retaining a correct diagnosis.
This work demonstrates the feasibility of using a GAN-based framework to accelerate ST MRF.Notably, while the in vivo human studies were focused on semi-solid MT imaging, only in vitro experiments were conducted for CEST.The main reason is that a recent work has shown that accurate in vivo CEST-MRF requires a more complicated imaging strategy, which involves a serial acquisition of T 1 , T 2 , B 0 , MT, and CEST data, and their integration in a sequential reconstruction pipeline. 34In particular the CEST reconstruction pipeline receives the MT reconstructed parameter maps as input.While the current study serves as the first demonstration that GAN-based architectures can be used to accelerate CEST-MRF, future work should be performed to validate its application in-vivo, which will require integrating the GAN architecture in a more complicated reconstruction pipeline.
Several additional steps could improve the GAN-ST performance.First, the training set used here was intentionally composed of healthy volunteers only, aimed to examine the extrapolation ability of the method.In addition, although 3D acquisitions create a large number of images, the number of training subjects scanned was relatively small (Table 1).Significantly increasing the training cohort, and training on a variety of pathological cases, are expected to boost the accuracy of the method.Second, the performance of the proposed method is dependent on the original discrimination ability of the M-length acquisition schedule, as only images from the end of the acquisition schedule can be removed, and not the beginning, due to the acquired spin history induced by earlier acquisitions.Therefore, the parameter discrimination ability could be further improved, while benefiting from the spatial denoising and extrapolation capabilities demonstrated here, by combining the proposed GAN approach with an optimized acquisition schedule, which could be discovered using recently developed deep-learning-based sequence optimization approaches. 63,64

CO N C LUS ION
The GAN-ST framework has demonstrated the ability to accelerate 3D acquisitions of semisolid MT and CEST mapping by 70% while maintaining excellent agreement with full-length CEST-MRF-based reference maps and retaining performance across unseen pathologies and scanner models.Furthermore, GAN-ST has shown improvements over CEST-MRF in regions with large susceptibility artifacts.GAN-ST has exhibited promising initial results in direct estimation of compound concentration and pH from MRF encoded images.

F I G U R E 2
Generative adversarial network (GAN)-saturation transfer (ST) in vitro image results.(A,B) L-arginine concentration (A) and exchange rate (B) maps from GAN-ST-based reconstruction, obtained with N = 9.Vials are numbered 1-6.(C,D) Full-length chemical exchange saturation transfer-magnetic resonance fingerprinting-based L-arginine concentration (C) and exchange rate (D) maps, obtained with M = 30.(E,F) GAN-ST-based (N = 9) concentration (E) and pH (F) maps.(G,H) Concentration (G) and pH (H) maps obtained using gold-standard non-MRI measures.F I G U R E 3 Statistical analysis and quantitative assessment of Generative adversarial network (GAN)-saturation transfer (ST) performance in vitro.(A,B) Correlation between GAN-ST based and chemical exchange saturation transfer-magnetic resonance fingerprinting-based concentration (A) and exchange rate (B) maps across the entire three-dimensional volume of an L-arginine phantom.(E,F) Box plots showing the distribution of per-vial GAN-ST-based L-arg concentration (E) and pH (F) maps with measured values indicated.Vial numbers are based on Figure 2A.(C,D,G,H) Structural similarity index and normalized root mean squared error for concentration/exchange rate (C,D) and concentration/pH (G,H) maps.

F I G U R E 4
Quantitative semisolid magnetization transfer (MT) parameter maps from a healthy volunteer, scanned at a site and scanner model that were not used during training.(A-D) Generative adversarial network (GAN)-saturation transfer (ST)-based semi-solid MT proton volume fraction maps, obtained with N = 9. (E-H) chemical exchange saturation transfer (CEST)-magnetic resonance fingerprinting (MRF)-based semisolid MT proton volume fraction maps, obtained with M = 30.(I-L) GAN-ST-based semi-solid MT proton exchange rate maps, obtained with N = 9. (M-P) CEST-MRF-based semi-solid MT proton exchange rate maps, obtained with M = 30.The red arrows indicate regions with susceptibility artifacts.
GAN-ST was able to accelerate the scan time by 70% as it required the acquisition of only N = 9 raw molecular information encoding images instead of M = 30.This translated F I G U R E 5 Quantitative semisolid magnetization transfer (MT) parameter maps from a glioblastoma patient.(A-D) Generative adversarial network (GAN)-saturation transfer (ST)-based semi-solid MT proton volume fraction maps, obtained with N = 9. (E-H) chemical exchange saturation transfer (CEST)-magnetic resonance fingerprinting (MRF)-based semisolid MT proton volume fraction maps, obtained with M = 30.(I-L) GAN-ST-based semi-solid MT proton exchange rate maps, obtained with N = 9. (M-P) CEST-MRF-based semi-solid MT proton exchange rate maps, obtained with M = 30.The red arrows indicate regions with susceptibility artifacts.into a CEST/MT protocol acquisition time of only 42-52 s, depending on the scanner model hardware and the number of coil channels.Moreover, for the brain imaging scenario, GAN-ST circumvented the need to acquire separate T 1 , T 2 , and B 0 maps, providing a total acceleration of about 91% (44 s instead of 8.5 min).The total training time was 4.66/8.71/12.81h, and the 3D inference time was 0.29/0.54/0.80 s, for the phantom, in vivo calf muscle, and in vivo brain, respectively.
, 30-40 min for each GBM patient monitoring session), it is essential to further accelerate semi-solid MT/CEST MRF, rendering it a cost-effective addition to clinical imaging protocols.The methodology proposed in this paper has made several contributions toward that cause.(1) The CEST-MRF saturation block was fused with a highly efficient and rapid snap-shot readout, allowing 3D semi-solid-MT or CEST-MRF acquisition in about 3 min, or 8.5 min (when B 0 , T 1 , and T 2 maps F I G U R E 6 Statistical analysis and quantitative assessment of the generative adversarial network (GAN)-saturation transfer (ST) performance in the in vivo brains of a tumor patient (A-D) and a healthy volunteer (E-H).(A,B) Correlation between all GAN-ST-based proton semi-solid magnetization transfer (MT) proton volume fractions (A) and exchange rates (B) for the entire brain in the WM/GM, and the corresponding pixel values obtained using chemical exchange saturation transfer (CEST)-magnetic resonance fingerprinting (MRF).Notably, the GAN-based f ss values in the WM are in better agreement with MRF refernce than the GM (Pearson's r = 90 compared to 0.74, respectively, p < 0.001), due to the myelin-rich content of the WM.(E,F) A similar analysis for the healthy human volunteer scanned at a site and scanner that were not available during training.(C,D,G,H) Structural similarity index metric and normalized root mean squared error for the tumor patient (C,D) and healthy volunteer (G,H).

39
15222594, 2023, 5, Downloaded from https://onlinelibrary.wiley.com/doi/10.1002/mrm.29574by Cochrane Israel, Wiley Online Library on [05/08/2023].See the Terms and Conditions (https://onlinelibrary.wiley.com/terms-and-conditions)on Wiley Online Library for rules of use; OA articles are governed by the applicable Creative Commons License Sevenfold data augmentation was performed using translations and horizontal/vertical image flips.The numbers mentioned in the table represent the acquired images prior to augmentation.